Automated Imaging Microscope System
The Aging Research Centre (ARC) is developing an Automated Imaging Microscope System (AIMS) to enable researchers to view the three-dimensional cellular organization of a region of tissue from a series of microscope slides. AIMS employs a computer-controlled microscope that can move a microscope slide along the X and Y axes, so that the field of view can be moved in equal increments. The microscope also has an autofocus mechanism. A CCD camera captures images, one field of view at a time. Using a magnification of 400 power, a typical slide contains 25,000 fields of view; perhaps 7000 of those fields contain something of interest. By using the thousands of individually captured images, the system is able to reconstruct a very high-resolution image of the entire slide.
AIMS analyzes each of the thousands of images and identifies and stores the following attributes for each cell it was instructed to identify: X,Y and Z coordinates for each desired cell type, amount of each stain in the cell, type of each stain in the cell, color of each stain in the cell and type of cell.
When AIMS has processed multiple layers of a tissue block, the system will be able to reconstruct a three-dimensional map of the cells in that block. In an experiment where there are two groups of animals and the researcher wishes to compare tissues from the two groups, AIMS will be able to compare three-dimensional structures as well as cell counts.
The system has two very different functions: acquisition and analysis. When we designed the system, we started on both ends simultaneously, attempting to analyze images captured from other sources, then working on our own data. Because we hope for the system to be used by biology researchers, not UNIX gurus, a decent user interface is one of the primary goals. We chose Tcl/Tk for the ease with which a GUI can be developed. The easily modifiable prototypes are useful. Various commands, written in C (but with Tcl/Tk interfaces) are used to control physical devices.
Physically, the system consists of an optical microscope, a CCD camera and three stepper motors hooked to the parallel port. The stepper motors move a microscope slide around the stage under computer control; basically, we have a 60,000bpi scanner.
The stepper motors are driven by Darlington-pair transistors (see http://www.doc.ic.ac.uk/~ih/doc/stepper/ for a discussion on stepper-motor control). This allows for great control, at the expense of hefty timing requirements. In the Linux kernel, we can find floppies driven every three to eight milliseconds. Empirically, given the load on our motors, we can sometimes move more than twice a second. We've hooked them up to the parallel port. The parallel port has eight data lines and four control lines which can be used for output. This is just sufficient to control three stepper motors at four controls/motor. If we needed more motors, we would have to use a different type of controller. Figure 1 shows the microscope looking at the focus mechanism. Figure 2 shows the circuit driving the motors. It uses two ULN2003A chips and is powered by a spare computer power supply.
We need to do a lot of moving, so speed is very important. Using nanosleep seems the simplest, if not the best, alternative. Combined with real-time priority, this causes the motors to move with a nice smooth hum. The other alternative, using the real-time clock at 2048KHz, doesn't allow as precise a control over speed. The major problem with this approach is the way nanosleep handles its delays. A busy-wait prevents any other task from running. RTLinux does seem like a better solution, although we haven't investigated that yet. The older usleep call is a poor choice, as it has 10ms granularity.
The greatest problem is figuring out the speed to move the motor. The speed at which a stepper motor is capable of moving depends on the load placed on it, and this load varies depending on the friction in the stage. A speed that works at one location may not work at all at another. Empirical experimentation seems necessary.
We can move in three dimensions. Not only can we view the entire slide, we can change the focus. A jury-rigged solution seems to work—move up and down, and pick the image with the most detail. We assume the one with the most detail is the most focused. Technically, “detail” is based on a “busyness” function. For each pixel, find the difference in intensity between each of its neighbors, then sum the absolute value of those differences.
Special Reports: DevOps
Have projects in development that need help? Have a great development operation in place that can ALWAYS be better? Regardless of where you are in your DevOps process, Linux Journal can help!
With deep focus on Collaborative Development, Continuous Testing and Release & Deployment, we offer here the DEFINITIVE DevOps for Dummies, a mobile Application Development Primer, advice & help from the experts, plus a host of other books, videos, podcasts and more. All free with a quick, one-time registration. Start browsing now...
- SUSE – “Will not diverge from its Open Source roots!”
- Dealing with Boundary Issues
- Vagrant Simplified
- Libreboot on an X60, Part I: the Setup
- System Status as SMS Text Messages
- Bluetooth Hacks
- October 2015 Issue of Linux Journal: Raspberry Pi
- Disney's Linux Light Bulbs (Not a "Luxo Jr." Reboot)
- New Products
- October 2015 Video Preview